[volt-nuts] Schematic error for Solartron 7081 (and 7071) in schematic 70817506 sheet 4 "AC Attentuator and Buffer"
David C. Partridge
david.partridge at perdrix.co.uk
Tue Sep 25 14:38:51 UTC 2012
C704 (22pf) is shown in the schematic as connecting from TP705 to the -15V rail.
This is not correct. It is actually connected from IC705 pin 6 to -15V (in other words between the base and collector of TR702).
More information about the volt-nuts