[volt-nuts] Solartron 7081 With Bad WR057 Diode
David C. Partridge
david.partridge at perdrix.co.uk
Fri Oct 31 06:45:37 EDT 2014
I might be worth asking eBay vendor LittleDiode, they are the eBay shop for LittleDiode Components ...
From: volt-nuts [mailto:volt-nuts-bounces at febo.com] On Behalf Of Charles Black
Sent: 31 October 2014 00:09
To: volt-nuts at febo.com
Subject: [volt-nuts] Solartron 7081 With Bad WR057 Diode
My Solartron 7081 has a bad WR057 diode that needs to be replaced. It's a three terminal device that looks like two silicon diodes in parallel.
It was making the meter read about 10.1V with a 10V input. Luckily it was temperature sensitive and was easy to find with freeze mist. Removal caused a shorted input to be about 2.5 microvolts high and a little noisy. I've not been able to find this device on the internet but thought that someone here might have seen this thing before.
volt-nuts mailing list -- volt-nuts at febo.com To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/volt-nuts
and follow the instructions there.
More information about the volt-nuts