[volt-nuts] Finally got around to modifying my Fluke 845ab with LED's
cfharris at erols.com
Tue Sep 9 16:42:30 EDT 2014
The thing is, CdS cells peak sharply at about 560nm, which is green/yellow.
They are blind to most of white light... especially to the blues which are
prominent in white LED's.
White LED's are designed purely for bulk illumination. They have a very
slow response so that they won't flicker when used for domestic lighting
off of a 50/60Hz power source.
There are also green yellow and red LED's that are designed the same way,
but there are many that switch quickly too.
Dallas Smith wrote:
> Hi Chuck,
> Its only 84Hz, It had the offset
> problem long before I made this mod. The new meter at work many decades ago
> also had this problem.
> I pick white so the full spectrum would
> be covered by the Csd's. Checked waveforms with scope, has sharp rising and
> falling edges. This mod may not be as elegant as a Linear Devices copper mod,
> but at lest I will not have to mess with the shielded front end.
> offset may be due to the bias current in the front end reacting with the different
More information about the volt-nuts