[volt-nuts] LTZ1000A Heater Temp (Ben Clark)
randyevans2688 at gmail.com
Fri Feb 13 22:05:12 EST 2015
Would you please post the link to your plot? I searched but haven't found
On Thu, Feb 12, 2015 at 3:51 AM, Jan Fredriksson <jan at 41hz.com> wrote:
> I posted a plot of calculated resistor values versus temperature in
> the LTZ1000 thread on the EVV blog yetserday. Email me if you can't
> find it.
> I guess the 3458A was specified to be used at north pole and desert
> heat, that's why they have such a high temperature setting. They could
> have a "laboratory version" and a "field version" of the reference....
> I have a NI-4071 PXI DMM with a LTZ1000 and it uses 14.3 ratio to
> fulfill the PXI specified temperature range. Still higher than optimal
> for stability but the temperature in my PXI rack is typically 40C at
> 20C room temperature.
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the volt-nuts