[volt-nuts] How accurate is a fluke after 9 years
Electronics and Books
electronicsandbooks at yahoo.com
Fri May 15 14:50:27 EDT 2015
My Fluke 5700A calibrator was calibrated today after 9 years. It displays how much it deviated from the last calibration. For the both voltage references i got 0.0 ppm and for the internal 10k resistor -1.7 ppm.
Only the amplifier for the 11-22 volt range drifted 1700 ppm for it depends on a voltage divider composed of two normal metalfilm resistors. These will be replaced by 2 vishay 0.01% resistors which are hopefully more stable.
Discere ne cesses
More information about the volt-nuts