[time-nuts] Morion MV89A Repair
lists at rtty.us
Sun Feb 26 16:20:16 UTC 2012
The paper mentions HC-40 crystals. They probably have the crystal drive current cranked down to optimize short term stability at 1 sec and longer. The phase noise floor takes a back seat.
The other interesting thing in the paper is that the short term numbers shown aren't into parts in the 13th.
On Feb 25, 2012, at 9:11 PM, "John Miles" <jmiles at pop.net> wrote:
>> Are the specs at 5 MHz or at 10? They should be 6 db worse at 10 than at
> Now that I'm looking back at the data sheet posted with the auction, they
> were indeed specified at 5 MHz. But even after the 6 dB penalty, their spec
> limits versus their real-world numbers are pretty marginal.
> Morion actually announced this particular oscillator at IFCS 2002, so
> there's a paper of sorts:
> http://www.ke5fx.com/Morion_MV89A_IEEE_IFCS_2002.pdf . Figure 4 was taken
> from a 10 MHz part, but all three of the ones I've tested look like the
> attached. Same as figure 4 until about 200 Hz... then they never go any
> -- john
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts