[time-nuts] PTS 3200 remote programming
rexa at sonic.net
Fri Mar 1 21:16:34 EST 2013
If you have a pdf manual, you probably have the same one I have. When I
was looking today, I did see that the input pin numbers listed in the
table, where it describes the programming (p 29 in the one I have or p32
of the pdf), does not match SCHEMATIC, PE-1121 (figure 7 or p44 of the
pdf). I think the schematic is not really for the 3200 as it doesn't
seem to show any inputs for the two GHz bits.
On 3/1/2013 4:37 PM, Iban Cardona wrote:
> Hi Alan,
> thanks for answer. Yes I have the manual, but the latches on my pts
> not are as are showed in the user manual. Maybe my unit it is a
> different version or release, or have some customozation from the pts
> In myunit, I tested the SO-2005 module that is that generates the
> 100Mhz decade, and I tested manually the 4bit to generate the 800Mhz
> and works well.
> My unit dont have front panel, only can be controllated remotelly.
> The differences from the user manual, is that un the user manual the
> 1ghz and 100mhz decade shares latch, and in my unit the shares latch
> the 100mhz with the 10mhz. The 1ghz latch is shared by the 1hz decade.
> Best regards
> 73! Iban
> On Fri, Mar 1, 2013 at 8:27 PM, Alan Melia<alan.melia at btinternet.com> wrote:
>> Hi Iban do you have a manual? they are available. From memory because its a
>> while since I worked on my Wavetek- Rockland unit I believe the remote
>> programming is in parallel with the front panel switches. The reason for
>> raising this is that there may be a problem on the back of the panel
>> switches.....mabe a wire off. Does the 800Mhz front panel switch setting
>> work correctly ??
>> ----- Original Message ----- From: "Iban Cardona"<icardona at gmail.com>
>> To:<time-nuts at febo.com>
>> Sent: Friday, March 01, 2013 6:42 PM
>> Subject: [time-nuts] PTS 3200 remote programming
>>> I'm traying to use to program the bcd interface of a PTS 3200.
>>> My truble us tha the 100Mhz decade 8 value bit is ignored by the PTS,
>>> then if I send 800Mhz to the PTS dont get output, and if send 900Mhz
>>> then I get 100Mhz.
>>> I checked the unit and the hardware looks OK. And my code is working
>>> well in another PTS units like the 620.
>>> The 3200 have some trick?
>>> Thanks for all
>>> 73! Iban
>>> time-nuts mailing list -- time-nuts at febo.com
>>> To unsubscribe, go to
>>> and follow the instructions there.
>> time-nuts mailing list -- time-nuts at febo.com
>> To unsubscribe, go to
>> and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts