[volt-nuts] Old HP3458A - SN: 2823A 03939
wpgold3637 at att.net
Sun Sep 21 12:04:37 EDT 2014
Push "blue" key then "blue R" ( the shifted "TRIG" key) and "ENTER".
Then press the downarrow "V" key 4 times to get to the "REV?" on the
display. Then press "ENTER" and two numbers will be displayed with a comma
in the middle. The first number is the multimeter's master processor
firmware revision. The second number is the slave processor's firmware
revision. This is if the "full menu" has been set. If not, see the "MENU"
command in the Operator's Manual and set it to "FULL".
Mine are both "7,2" and I have no problems. The latest is "9".
----- Original Message -----
From: "cfo" <xnews5 at luna.dyndns.dk>
To: <volt-nuts at febo.com>
Sent: Sunday, September 21, 2014 8:22 AM
Subject: [volt-nuts] Old HP3458A - SN: 2823A 03939
> I'm looking for a 3458A , and have found one : SN: 2823A 03939 w. opt 002
> Seller says it's passing selftest.
> But it's an old serial# , is there any way one can verify that it has an
> up-to date firmware , and maybe via that see if it has has all the needed
> upgrades or ServiceNotes applied ?
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the volt-nuts